Intra-Oral Scan Bodies
Scanning by laser, contact or imaging techniques requires an extended scan body, clear rotational features and a surface that is scannable without applying any powder or colouring. The connection to the mating part, i.e. implant, requires high quality machining and precision.
Neoss® Scan Bodies are designed and manufactured to ensure high and consistent accuracy of the digital impression. Scan bodies are available for Neoss implant platforms SP (Ø3.5-6.5 implants) and NP (Ø3.25 implant) and Access abutment level. Each scan body is marked for ease of selection. Neoss Scan Bodies (configurations with SP, NP and Access marking on the product) are compatible with the Neo screwdriver but also the original Scan Body hex screwdriver.
Scan bodies are compatible with most available scanners and planning and design software including 3shape, Exocad and Dental Wings.
Simply download the appropriate Neoss Library available from Neoss.
Neoss Scan Bodies are placed directly on implants or abutments in the patients’ mouth and are intended as a means to perform a digital impression using an intra oral scanner. Alternatively these can be placed on an implant replica in a model and a digital impression can be recorded using a desktop 3D scanner.
The purpose of a digital impression is to accurately transfer the position of the implant, or sometimes an abutment, in relation to natural teeth, other implants and soft tissue contours to a digital laboratory model in preparation for design work of a restorative solution using different design software.
A digital impression may be taken at different stages during treatment dependent on the operator preferences and can be taken both intra-orally or at the dental laboratory:
- At time of initial surgery – for one stage techniques, or to enable the delivery of a provisional crown at second stage surgery
- At second stage surgery - following soft tissue healing after a second stage surgical procedure
- At the dental laboratory - following a conventional impression at either of the stages above
Note: For digital impressions of more than one implant, each implant/abutment requires a scan body during the scanning. In addition for larger cases only desktop scanners or dedicated scan centers are recommended to achieve the appropriate accuracy of the digital impression.
For all procedure details please see instructions for use.
The Neoss scan bodies are only to be used with the Neo screwdriver or the original Scan Body hex screw driver and original Neoss digital libraries.
See Product Catalog for order details.
PEEK (natural white) with a titanium base and screw.
Scan bodies and related screwdriver are available as individual items.